LIF thermometry in a thermoacoustic device Courtesy: Jaworski , Uni. Manchester, MST 21 (2010)
LIF Imaging in Thermal Flows
In LaVision’sFluidMastersystems for LIF-Thermometry in thermal flows a variety of temperature sensitive LIF tracers are used.
At constant tracer seeding density the LIF signal is directly related to the local flow temperature, but laser light sheet and laser beam absorption corrections has to be applied. These corrections are provided in our LIF image processing module inDaVis.
Related application: Laser imaging of gas temperature w/o flow seeding: FluidMasterlaser imaging system based on Rayleigh thermometry
LIF-thermometry and PIV in a convective water flow
In case the LIF tracer concentration is not constant, 2-color LIF thermometry is the best choice for fluid temperature measurements. This LIF imaging method uses the image ratio of two LIF emissions at different wavelengths (colors) to measure the fluid temperature. 2-color LIF ratio imaging is not only independent of the local tracer seeding concentration but also insensitive to changing illumination conditions across the laser light sheet. This superior image ratio approach needs a second camera or an image doubler in front of the lens of a single camera.
LaVision’sDaVissoftware is supporting multi-camera imaging with sophisticated image mapping functions for multi-directional views.